Chemists using silica columns for normal-phase flash chromatography typically equilibrate their columns prior to loading their samples. Companies manufacturing automated flash purification systems often have the equilibration volume and flow rate pre-programmed and tied to a column size. Some of these flash system companies allow equilibration volume to be edited while others have the volume fixed. Is one of these options better than the others? In this post I discuss how equilibration volume impacts flash chromatography results.
Flash chromatography is a standard part of an organic chemist’s workflow. It is utilized after most reaction steps in order to remove most of the generated by-products and excess reagents.
This is an interesting question that I am asked from time to time. There does seem to be two camps in which chemists reside – one believing longer and thinner columns provide better separations and the other preferring shorter and fatter columns to do the same chromatography.
Which is right? That is a question I will try to answer based on my own data.
Recently, I posted an article explaining why high performance TLC plates are not needed for method development for high-performance flash chromatography. Based on some excellent feedback, I see a need to discuss silica chemistry and its impact on chromatography.
Have you ever run flash column chromatography with mass detection (Flash-MS) and observed the total ion current or TIC increase during the purification only to find that there was no discernible compound contributing to the effect?
In this post I discuss how I came across this issue and the solution I found to work.
The products of organic synthesis are designed with specific functional groups in order to possess desired properties. Depending on the compound’s functionality, it can be neutral, acidic, or basic as determined by a compound measurement called pKa or acid dissociation constant. Compounds with low pKa are typically acidic while those with high pKa tend to be basic. Compounds with a pKa near 7 are deemed neutral.
In my previous post, I talked about the "Chemistry Behind Normal-phase Flash Chromatography", the most common form of liquid-solid chromatography. In this post, I focus on reversed-phase flash chromatography and how it differs from normal-phase.
The challenges organic, medicinal, and natural product chemists face are many: from designing reactions, to optimizing synthesis, work-up / extraction, and purification / isolation of the desired compound or compounds. Among those issues related to purification / isolation is the common problem of separating compounds with similar chemistry that either co-elute or separate poorly.
In this post I will discuss some tips on how to "resolve" this issue (yes, pun intended).
In a previous post I talked about column size, specifically long-thin versus short-fat and the impact of the cartridge’s dimensions on purification performance. With that comparison I showed that in preparative chromatography, purification efficiency is more about the amount of silica than column dimensions. Cartridges of different dimensions containing the same amount of the same media will provide the same separation efficiency.
A question I hear a lot from chemists is “how much can I load”. The answer is always “it depends on your separation quality”. At that point I begin asking about the TLC data and purification goals. Purification goal setting should be your first step and the question to answer is – what do I need this purification to achieve? Is the goal high purity, high yield, or some combination. Remember, you will typically sacrifice purity for high yield and yield for high purity so optimization is an important consideration.
Over the past several decades, the chemical industry has implemented process changes and updated practices in R&D and manufacturing in an effort to reduce liquid and solid lab waste. The pharmaceutical industry in particular has taken steps within their drug discovery labs to reduce solvent use by requiring their chemists to find and implement measures that achieve the corporate environmental goals without curtailing their productivity – quite the challenge.
Flash chromatography – a purification tool for both organic chemists and natural product researchers. This tool is essential when you need to remove impurities from your targeted product, or products, in order to get them pure. To reduce the costs associated with flash chromatography, some chemists try reusing the same column over and over, not always with success.
Synthetic organic chemistry is the genesis of new pharmaceutical and commercial chemical products. In short, it is based on the idea that two or more carbon-based compounds can be forced to react using heat, or other energy source, to create a new, novel product – but this we already know.
For chemists preferring or needing to dry load their crude sample mixtures to get an acceptable flash purification result, using the right ratio of sample to sorbent can be quite important. Too much sample and solubility issues can ensue, too little sample and significant band broadening occurs, reducing the separation quality.
In this post, I propose an acceptable ratio range based on my own experimental data.
When Isolera™ was launched, the maximum system pressure that could be reached was 10 bars, but reaching that pressure was a challenge since most of the Flash columns could not withstand the higher pressures. The maximum pressure rating for the Biotage® SNAP columns, for example, is limited to five or seven bars, depending on the size, and columns from most of manufacturers have the same limitation.
On a scenic drive up the I-15 in southern California, I got to take a tour of the undergraduate lab at California State University, San Marcos with Dr. Robert Iafe. His lab is one of the first to have the new Biotage® Selekt Flash Purification System paired with the Sfär columns. We discussed the impact on his undergraduates broadening capabilities to learn about new instruments, and how it has affected his own research in his goals to gain tenure at the university.